VCLAS (Visualase Cooled Laser Ablation System)   Part Number: 9735559 Visualase Cooled Laser Applicator System 3mm Tip;  Part Number: 9735560 Visualase Cooled Laser Applicator System 10mm Tip;  Part Number: 9735561 Visualase Cooled Laser Applicator System 15mm Tip
The VCLAS 3mm, 10mm, and 15mm tip devices were not appropriately tested for saline leakage during the manufacturing process.
Worldwide Distribution - USA (nationwide) and to the countries of :  Canada, Australia, UAE, Kuwait, Saudi Arabia, South Africa
Alaris Pump Module model 8100 manufactured between November 2011 and March 2012;   Alaris Pump Module serviced with LVP Mechanism Sub Assembly (P/N) 10942012 between November 2011 and March 2012;   Alaris Pump module Bezel Kit Assembly (P/N) 10964559) shipped between November 2011 and March 2012.      The Alaris Pump module is a large volume infusion pump offered under the Alaris System. The Alaris Pump module will deliver medication and fluids using the IV administration sets for continuous or intermittent delivery through clinically acceptable routes of administration such as intravenous or epidural.
The recalling firm has received reports of increased or decreased flows that have occurred in certain pumps.
Merlin@Home Software Model EX2000 v8.2.2 for Merlin@Home Transmitter Models EX1150, EX1150W, EX1100, and EX1100W.  The Merlin@home Transmitters are a component of St. Jude Medicals (SJM) Remote Care System.  These Merlin@home transmitters are used to retrieve information from the SJM implantable medical device (IMD) using RF telemetry.  The data is then sent back to the Merlin.net Patient Care Network through a built-in modem or, alternatively, through a USB cellular adapter or a USB based Broadband Kit.  The transmitted data enables clinicians to remotely monitor a patients IMD using a secure, encrypted, password-protected web site, thus reducing the number of in-clinic visits a patient needs to make to their physician.  The transmitter monitors the patients device daily between scheduled follow-ups and can alert a clinician if it detects anything out of the norm.
New pacemaker firmware was developed to further mitigate the risk of unauthorized access to our pacemakers that utilize radio frequency (RF) communications.
Assurity Model Numbers: PM1240, PM2240; Assurity + Model Number: PM2260; Accent Model Numbers: PM1210, PM2210, PM1222, PM2212; Accent MRI Model Numbers: PM1224, PM2218, PM2224  Implantation of a single-chamber pulse generator, dual-chamber pulse generator, or CRT-P is indicated in one or more of the following permanent conditions: Syncope, Presyncope, Fatigue, Disorientation, Or any combination of those symptoms. Implantation of a CRT-P is indicated for: Maintaining synchrony of the left and right ventricles in patients who have undergone an AV nodal ablation for chronic atrial fibrillation and have NYHA Class II or III heart failure. The reduction of the symptoms of moderate to severe heart failure (NYHA Class III or IV) in those patients who remain symptomatic despite stable, optimal medical therapy, and have a left ventricular ejection fraction d35% and a prolonged QRS duration
PM1226 ACCENT ST MRI SR RF and PM2222 ACCENT ST  These low voltage (LV) devices are implantable pacemaker pulse generators, intended to be permanently implanted in the body, that have a power supply and electronic circuits that produce a periodic electrical pulse to stimulate the heart.  These devices are used as a substitute for the heart's intrinsic pacing system to correct both intermittent and continuous cardiac rhythm disorders.
Anthem Model Numbers: PM3210, PM3212; Allure/Relieve Model Numbers: PM3222, PM3242, PM3262, PM3542, PM3224, PM3244  Implantation of a single-chamber pulse generator, dual-chamber pulse generator, or CRT-P is indicated in one or more of the following permanent conditions: Syncope, Presyncope, Fatigue, Disorientation, Or any combination of those symptoms. Implantation of a CRT-P is indicated for: Maintaining synchrony of the left and right ventricles in patients who have undergone an AV nodal ablation for chronic atrial fibrillation and have NYHA Class II or III heart failure. The reduction of the symptoms of moderate to severe heart failure (NYHA Class III or IV) in those patients who remain symptomatic despite stable, optimal medical therapy, and have a left ventricular ejection fraction d35% and a prolonged QRS duration
PM3544 QUADRA RELIEVE and PM3562 QUADRA ALLURE MP  These low voltage (LV) devices are implantable pacemaker pulse generators, intended to be permanently implanted in the body, that have a power supply and electronic circuits that produce a periodic electrical pulse to stimulate the heart. These devices are used as a substitute for the heart's intrinsic pacing system to correct both intermittent and continuous cardiac rhythm disorders
PM 1250 and PM 2250 ZENEX; PM1282 and PM2282 ZENEX  MRI  These low voltage (LV) devices are implantable pacemaker pulse generators, intended to be permanently implanted in the body, that have a power supply and electronic circuits that produce a periodic electrical pulse to stimulate the heart.  These devices are used as a substitute for the heart's intrinsic pacing system to correct both intermittent and continuous cardiac rhythm disorders.
PM 1260 ASSURITY +  These low voltage (LV) devices are implantable pacemaker pulse generators, intended to be permanently implanted in the body, that have a power supply and electronic circuits that produce a periodic electrical pulse to stimulate the heart.  These devices are used as a substitute for the heart's intrinsic pacing system to correct both intermittent and continuous cardiac rhythm disorders.
Penrose-St Francis Health Services Blood Bank
Apheresis Platelets, Leukocytes Reduced, Irradiated
Blood product, for which quality control and distribution of product did not meet specifications, was distributed.
Death Wish Nitro Canned Cold Brew Coffee 11 oz  UPC: 851552005599
Blood product, collected from a donor whose suitability to donate was not adequately determined, was distributed.
Blood product, collected from a donor who traveled to a malarial endemic area, was distributed.
Philips Electronics North  America Corporation
Philips Healthcare Brilliance 64 System, X-Ray, Tomography, Computed    These computed tomography x-ray systems are intended to produce cross-sectional images of the body by computer reconstruction of x-ray transmission data taken at different angles and planes.
During a bolus tracking procedure, no images were generated when the scan was completed, and the raw data file was not available for offline reconstruction. There is a discrepancy between the calculated reconstruction length and the actual scan length that results in the inability to reconstruct raw data. The operator may choose to rescan the patient.
Philips Healthcare  Ingenuity  Core128   X-Ray, Tomography, Computed      These computed tomography x-ray systems are intended to produce cross-sectional images of the body by computer reconstruction of x-ray transmission data taken at different angles and planes.
Philips Healthcare Ingenuity Core  X-Ray, Tomography, Computed    These computed tomography x-ray systems are intended to produce cross-sectional images of the body by computer reconstruction of x-ray transmission data taken at different angles and planes.
Philips Healthcare Ingenuity CT   X-Ray, Tomography, Computed      These computed tomography x-ray systems are intended to produce cross-sectional images of the body by computer reconstruction of x-ray transmission data taken at different angles and planes.
Philips Healthcare Brilliance iCT   X-Ray, Tomography, Computed      These computed tomography x-ray systems are intended to produce cross-sectional images of the body by computer reconstruction of x-ray transmission data taken at different angles and planes.
Philips Healthcare Brilliance iCT SP  X-Ray, Tomography, Computed      These computed tomography x-ray systems are intended to produce cross-sectional images of the body by computer reconstruction of x-ray transmission data taken at different angles and planes.
Worldwide Distribution - US (nationwide) (foreign consignee list will be provide by the firm at a later time)
Proclaim DRG Implantable Pulse Generator Model 3664
Analysis of the difference between the actual device longevity and the battery indicator status provided by the Clinician Programmer (CP) or Patient Controller (PC) has revealed errors in the CP or PC longevity estimations that can overestimate or under estimate the battery indicator status in certain situations.
Infinity 5 Implantable Pulse Generator and Infinity 7 Implantable Pulse Generator Models 6660, 6661, 6662, 6663
Proclaim 5 Elite Implantable Pulse Generator and Proclaim 7 Elite Implantable Pulse Generator Models 3660, 3661, 3662, 3663, 3665, 3667
Alprazolam Tablets, USP, 0.25 mg, 100-count Unit Dose tablets per carton, Rx Only, Manufactured by: Watson Pharma Private Limited, Verna, Salcette Goa 403 722 INDIA; Distributed by: Actavis Pharma, Inc., Parsippany, NJ  07054; Distributed by: Major Pharmaceuticals, 17177 N. Laurel Park Dr., Suite 233, Livonia, MI 48152, NDC 0904-5858-61.
Failed Impurities/Degradation Specifications: Out-of-specification result for an unknown impurity.
Blood products, which did not meet the acceptable product specifications, were distributed.
Two or more of the following: Email, Fax, Letter, Press Release, Telephone, Visit
Dia-Ionx, 4 fl oz (118 mL) plastic bottle.  Distributed by VETiONX, 4635 40th St. SE, Grand Rapids, MI 49512.  NDC:  50364-003-04
Presence of mold and bacteria in veterinary products.
Arthro-Ionx, 4 fl oz (118 mL) plastic bottle.  Distributed by VETiONX, 4635 40th St. SE, Grand Rapids, MI 49512.  NDC:  50364-014-04
Promaxol, [4oz or 32 oz] plastic bottle.  Distributed by VETiONX, 4635 40th St. SE, Grand Rapids, MI 49512.  NDC:  50364-013-04 (4 oz) and 50364-013-32 (32 oz)
Lot Numbers:   Pro/030617A (4oz), Lot Number Pro/030617B (32oz)
Toshiba Medical Kalare Fluoroscopic X-Ray System (DREX-KL80)       Kalare is intended to be used as a universal diagnostic imaging system for radiographic and fluoroscopic examinations, including general R&#38;F and pediatric examinations. It is intended for use by qualified/trained doctors or technologists on both adult and pediatric subjects taking radiographic and fluoroscopic exposures of the whole body, skull, spinal column, chest, abdomen, extremities, and other body parts. Applications can be performed with the patient sitting, standing or lying in prone or supine positions.
During an examination images were displayed on the live monitor,but the images were not displayed on the system monitor nor were they saved to the hard disk.
Product was distributed throughout the United States.
Symbio Muc Eye, 0.17 fl.oz. 5mL bottles, Manufactured for and distributed in the US b: Biological Health Group Corp. Bremerton, WA 98337. NDC 69711-125-61
methylPREDNIsolone ACETATE, 40 MG/ML INJECTION SUSPENSION, 10 ML Vial, RXQ Compounding, Athens, OH --- NDC:  70731-0936-10
Lack of Assurance of Sterility; autoclave parameters were not correct for one of two autoclaves used to sterilize the product.
USA, Austria, Australia, Belgium, Brazil, Canada, Chile, Colombia, Germany, Spain, France, UK, Ireland, Israel, Italy, South Korea, Lebanon, Netherlands, Philippines, Saudi Arabia, Sweden, Uruguay
Xpert EV Assay, Catalog GXEV-100N-10.    The Cepheid Xpert EV assay is a reverse transcription polymerase chain reaction (RT-PCR) using the GeneXpert Dx System for the presumptive qualitative detection of enterovirus (EV) RNA in cerebrospinal fluid (CSF) specimens from individuals with signs and symptoms of meningitis.
Customers have reported a higher than expected number of invalid test results when using the recalled product per the package insert.
Product was shipped to the following states: CO, NM, UT & WY.
Chicken Salad Sandwich 6.8 oz plastic container Contains: Wheat, Egg, Soy  USPC 6 63209-02050 9
The firm was notified by their manufacturer that the product is contaminated with Listeria monocytogenes.
US Distribution to the states of : FL, TX, NY, IL, GA, AL, MA, MO
Genesys spine, Binary Screwdriver, GCP221, component of the Binary Anterior Cervical Plating System      The Genesys Spine Binary Anterior Cervical Plate System is intended for anterior screw fixation to the cervical spine. It is to be used in skeletally mature patients as an adjunct to fusion of the cervical spine (C2 to C7). The system is indicated for use in the temporary stabilization of the anterior spine during the development of cervical spinal fusion in patients with degenerative disc disease (as defined by neck pain of discogenic origin with degeneration of the disc confirmed by patient history and radiographic studies), spondylolisthesis, trauma (i.e. fractures or dislocations), tumors, deformity (defined as kyphosis, lordosis, or scoliosis),pseudoarthrosis, failed previous fusion, and/or spinal stenosis.
Unanticipated technique for a cervical screw insertion caused the tip of the threaded inner shaft to break off.
In this situation, donor gave history which warranted deferral or follow up and was not deferred or follow up questions was not asked, was distributed.
Tampico Tropical Punch packaged in 1 gallon plastic jugs
Worldwide Distribution: US (Nationwide) and countries of: Canada, France, Taiwan, Argentina, Australia, S. Korea, Belgium, Singapore, India, Indonesia, Philippines and Thailand.
Pilling(R) Knife Handles:  a) REF 352950   b) REF 352953   c) REF 352951   d) REF 352957   e) REF 352952 :     Reusable surgical knife handles, which are intended for use with blades which are inserted into a slot in the handle.
The knife handles slot depth is out of specification, so blades may not fit on the handles properly.
a) REF 352950: Lot numbers: D6, J6, E6, F6, G6, H6, I6  b) REF 352953: Lot numbers:  I6, B6, E6   c) REF 352951:  Lot numbers: I6, J6  d) REF 352957:  Lot numbers: A6, B6, B7, K6, L6  e) REF 352952:  Lot numbers: H6
Panther(R) System, Catalog Number 902615, For Use with Hologic Assays, For In Vitro Diagnostic Use
An incorrect incubator was installed on one Panther System which potentially led to incorrect Aptima Combo 2(R) assay (AC2) results.
Bisoprolol Fumarate tablets,  5 mg,  30-count bottles, Rx only, Manufactured by: UNICHEM LABORATORIES, LTD. Pilerne Ind. Estate, Pilerne, Bardez, Goa, 403 511, India, Marketed by: UNICHEM PHARMACEUTICALS, (USA), INC. Hasbrouck Heights, NJ 07604, NDC 29300-126-13.
Failed Impurities/Degradation Specifications: Out of Specification for an unknown impurity  was observed during the 18 month stability testing.
AL, KS, PA, FL, ID, OR, IA, TX, AR, CA, MT, KY, WA
Corid (Amprolium), 9.6% Oral Solution, Coccidiostat.  Net Contents:  1 gal. (128 fl. oz) (3,785 mL).  Distributed By:  Merial Limited, 3239 Satellite Blvd., Duluth, GA 30096-4640, USA.  NADA: 013-149
Confirmed out of specification for a low assay value on one sample at the 6-month stability time point.
Merkts Port Wine Cheese Spread, Net Wt. 14 oz  (396g), UPC 76306-60200
Bel Brands USA is recalling Merkts Port Wine Cheese Spread after being notified by customers that white plastic pieces were found in the product.
U.S. distribution to the following: CA, NJ, CO, GA, FL, MD, PA, MA, IN, WI, TX, and NY.    No foreign distribution.
La Terra Fina  Spinach Artichoke & Parmesan   Dip & Spread  NET WT. 10 oz. (.625 lb.) 284g     UPC 6-40410-51327-3
Lot number: 20170912  Use By Date: NOV-01-2017
LifeSouth Community Blood Centers Inc - Birmingham Region
Apheresis product, in which platelet yield did not meet specifications, was distributed.
Blood Bank Of San Bernardino And Riverside Counties
Blood products, collected from a donor for whom donor suitability was not adequately determined, were distributed.
Blood and Blood Products for Reprocessing
Misbranded red blood cell, was distributed.
Source Plasma units, not meeting product specifications, were distributed.
Source Plasma unit, not meeting product specifications, was distributed.
Human cornea, recovered from a donor whose donor eligibility determination failed to ensure that the donor was free from risk factors for, and clinical evidence of infection due to relevant communicable disease agents and diseases, was distributed.